January 25, 2006, Cincinnati, OH – THE MODAL SHOP, INC., a PCB Group Co., introduces the addition of sensor linearity check capabilities to the workhorse Model 9155C Accelerometer Calibration Workstation.
The Model 9155C-501 Linearity option allows users to perform multipoint sensor linearity checks up to 40gpk using The Modal Shop’s calibration-grade air-bearing shaker. Linearity checks can be performed up to an amplitude of 500gpk with a mechanical amplifier bar. Verifying linearity provides additional assurance of sensor health and performance, increasing confidence in measurement accuracy.
The easy-to-use software GUI automates data acquisition across specified amplitude range and seamlessly integrates into The Modal Shop’s Model 9155C Accelerometer Calibration Workstation that performs calibrations on both ICP® (IEPE) and charge mode accelerometers, providing both sensitivity and phase data. Calibrations are performed using the 'back-to-back' method according to ISO 16063-21. All sensor information and calibration data is managed within a Microsoft Access database that allows for customizable calibration reports conforming to the requirements of ISO 17025 for calibration certificates. The system also self-documents measurement uncertainty according to ISO 16063-21.
For more information on the Model 9155C Accelerometer Calibration Workstation or other sound and vibration sensor calibration products, contact THE MODAL SHOP, INC., 3149 E. Kemper Road Cincinnati, OH 45241-1516, Internet: www.modalshop.com, (800) 860-4867, (513) 351-9919, Fax (513) 458-2172 or e-mail: firstname.lastname@example.org.
# # #
ICP is a registered trademark of PCB Group Inc.